$525.00 $799.00 You Save 34% ($274.00)
Manufacturer: Bovie Medical | Manufacturer Part #: D101
Warranty: 4 year manufacturer
The Derm 101 High Frequency Desiccator offers ten watts of monopolar power, adjustable in 1/10th watt increments, to desiccate and fulgurate tissue. The power can be adjusted using the 3-Button Power Control Handpiece, A902, (Patent #8,998,899) or using the rotary dial on the unit. The unit features a large, bright, LED display for excellent visibility across the procedure room. If foot control activation is preferred while using the DermMedStockUSA.com 101, just plug in the A803 Bovie Footswitch (sold separately) and activate. No separate handpiece is required.
For the practitioner who wants to also use bi-polar electrosurgery along with high frequency desiccation, Bovie offers the Derm 102 High Frequency Desiccator with 10 watts of bipolar power, adjustable in 1/10th watt increments. To utilize this feature the A827V* bipolar cable, a bipolar forcep* and footswitch activation are required.
The Derm 101 and Derm 102 are packaged ready to use. The units come with a total of 18 disposable dermal tips. By the time the practitioners utilizes all 18 electrodes, they will have paid for the unit, twice, and generated income for their practice.
|FEATURES||DERM 101||DERM 102|
|Bipolar (utilizes A827V* bipolar cable)||X|
|1/10th Watt Power Adjustment||X||X|
|Foot Control Setup(utilizes A803* footswitch sold separately)||1 step||1 step|
|Foot Control Handpiece Required||No||No|
INCLUDED IN THE BOX:
A) POWER SWITCH
Turn the unit power on utilizing the switch on the right side panel of the unit.
B) POWER OUTPUT DIAL
Set the power output either by using the dial on the front of the unit (B) or on Bovie DERM 101/102 unit only by the up and down buttons on the handpiece (F). When power level adjustment is being made by the handpiece an audible tone will sound to indicate that the power level has been changed. Depressing and holding the up or down buttons will cause the power settings to change more rapidly for quick adjustment of the output settings. Power output is displayed in “.1” watt increments from 0.1 to 10 watts.
NOTICE: The output settings can not be adjusted when the unit is being activated.
C) ACTIVE LIGHT
When the unit is activated, an audible tone is sounded and the blue active light will illuminate.
D) STANDARD ELECTRODE
Slide the standard electrode into the handpiece until it is firmly seated. The handpiece will accept most standard 3/32″ electrodes.
E) ACTIVATION BUTTON
To activate the unit, remove the handpiece from the holder. Place the handpiece in the desired position and depress the activation button.
F) POWER OUTPUT BUTTONS
Set the power output by pressing the up and down buttons on the handpiece.
G) MONOPOLAR OUTPUT RECEPTACLE
The monopolar output for the handpiece is on the lower left front of the unit. The handpiece plug is designed to fit in only one direction. Plug the connector from the handpiece into the receptacle on the bottom of the unit. The three button handpiece is designed to give the doctor complete fingertip control of the power output settings.
H) GROUNDING PLATE OUTPUT RECEPTACLE
To use the optional grounding plate with cord (A802EU), insert the plug of the cord into the grounding plate output and connect the other end into the grounding plate. The plate should be placed underneath the patient at a point where the entire plate is covered by bare skin. The use of conductive gel is recommended.
I) BIPOLAR OUTPUT RECEPTACLE (DERM 102 only)
To use the optional bipolar cord (A827V), insert the pins into the bipolar outputs. The cord is then plugged into the forceps. A sliding gate behind the monopolar and bipolar outputs prevents the user from using both simultaneously.
J) POWER CORD
Plug the female end of the power cord into the base of the unit.
K) FOOTSWITCH OUTPUT RECEPTACLE
The optional footswitch (A803) is plugged into the footswitch output and placed on the floor. The footswitch can be used with monopolar procedures and must be used with bipolar procedures.
|CREST FACTOR @
|Coagulation||10 W @ 1000Ω||550 kHz ± 44.9 kHz||19.5 kHz ± 10%||10.0 ± 20%||3.3 kV|
|Bipolar||10 W @ 150Ω||550 kHz ± 44.9 kHz||19.5 kHz ± 10%||10.0 ± 20%||1.0 kV|
Main Voltage: 100 – 240 VAC ± 10%
Main Frequency: 50 – 60 Hertz
Main Current: 320mA Max.
Power Consumption: 28 VA
Height 8.2” (208mm)
Width 7.2” (184mm)
Depth 2.7” (69mm)
< 1.4 kg (< 3 lbs)
Basic Construction: In accordance with
Mode of Operation: Intermittent Operation
Output Type: TYPE BF
Duty Cycle: 10sec on / 30sec off
Main Fuses: T 400mAL, 250V
Be the first to ask a question about this.